Cleaning machine is the production, sales, maintenance of ultrasonic cleaning machine for the whole company, the development of the company technical force is constantly in recent years, technology innovation and production, products are sold throughout the country, by the customer the consistent high praise, let everyone know about ultrasonic cleaning tank design.
Ultrasonic cleaning groove by groove and an outer shell, the outer surface of the inner groove (general appearance at the bottom of the groove) bonding ultrasonic transducer, groove Sheng cleaning fluid. Groove is generally made of stainless steel plate corrosion, too thick will affect the ultrasonic radiation, cell wall, especially the radiation plate adhered with transducer to smooth polished, can not have scar, or it is easy to produce cavitation corrosion, shorten the service life. To avoid direct contact with the workpiece cleaning tank wall plate and scratch, hollow basket with general (cleaning basket) or stent will clean the suspension in the cleaning solution. The basket frame should be small and light, generally with the stainless steel wire plaited or made of other reflected sound good material. The structure to make the ultrasonic cleaning liquid flows easily blocked small. The inner tank size according to the size and shape of the cleaning element set. The total surface area of cleaning of large should be greater than the groove volume. Stick with a radiation plate transducer (such as the tank bottom plate) power strength under generally lower than 1.5W? CM2) (voltage transducer, most should be between 0.5-1w/cm2). High strength will accelerate cavitation corrosion radiation plate surface, at the same time due to severe cavitation bubbles generated will affect the energy transfer, so far away from the radiation surface liquid space sound intensity becomes weak and uneven cleaning purpose. In the cleaning tank in general, because the level of reflection, in the cleaning tank will produce a standing wave, the liquid space some regional pressure minimum (nodes), some local pressure maximum (antinode) caused by uneven clean degree. In order to reduce the formation of standing wave cleaning trough, sometimes the shape to be specially designed, or take other measures, such as sweep mode. Cleaning items in the groove are arranged to have a certain interval, and the narrow surface radiation toward the transducer, so as not to impede the sound radiation into the cleaning groove space. For more information please visit the ultrasonic cleaning tank cleaning machine: www.dsedh.com.